Pressure measured using vacuum as reference. Usually
expressed in psia.
The pressure difference between two pressure
sources, which usually are neither vacuum nor the ambient
pressure. Usually expressed in psid.
Pressure measured using ambient pressure as reference.
Usually expressed in psig.
Sealed Gage Pressure:
Pressure measured using a fixed pressure as reference. The fixed pressure is usually 1 atm (14.697 psia).
Usually expressed in psis.
Pressure of the air surrounding the earth. This pressure varies depending on altitude and
see atmospheric pressure
Pressure surrounding the device.
Average temperature surrounding the device.
A voltage or current signal that is a continuous function of the measured parameter.
American National Standards Institute
National Electrical Manufacturers Association
The specified range over which a transducer is intended to measure,
specified by the upper and lower limits.
Compensated temperature range:
Temperature range in which a sensor meets all claimed specifications.
Operating temperature range:
Temperature range in which a sensor can operate without damage. But specifications claimed
may not be met in this range.
Storage temperature range:
Temperature range in which sensor can be safely stored. At these temperatures, sensor is not supposed
to be power-supplied or to operate.
Material added to a device to reduce a known error.
The arithmetic difference in transducer output signal measured at the specified minimum and
maximum operating pressures.
The ratio of output signal voltage change to the corresponding input pressure change.
Sensitivity is determined by computing the ratio of span to the specified input pressure range.
The temperature used as reference in measuring transducer errors.
The algebraic difference between endpoints. Where one endpoint is actual offset voltage and
the other endpoint is the upper limit of the range.
The pressure used as a reference in measuring transducer errors.
The transducer output signal obtained when the reference pressure is applied.
The maximum normal mode (measured) pressure that can be applied without changing the
transducer's performance or accuracy beyond the specified limits. This would be
applied to either pod of a differential transducer. This is also called PROOF PRESSURE.
The maximum pressure that can be applied to both ports simultaneously of a differential transducer.
The maximum pressure that can be applied to a transducer without rupture of
Best Straight Line (BSL):
The best straight line chosen such that the true transducer response curve
contains three points of equal maximum deviation.
Temperature Coefficient (TC):
The error band resulting from maximum deviation of a transducer output
parameter (such as offset or span) as temperature is varied from 25ºC to any
other temperature within the specified range. It is usually measured in ppm/ºC or µV/V/ºC.
The error band expressing the ability of the transducer to reproduce an output signal
parameter (such as offset or span), at specified pressures and temperature,
after exposure to any otter pressure and temperature within the specified range.
The error band expressing the ability of a transducer to maintain the value of an output
parameter (such as offset or span) with constant temperature and pressure inputs.
The error band defined by the maximum signal deviation obtained when a transducer is
replaced by any other transducer of the same type with equivalent pressure
inputs and temperature ranges.
Most Probable Error:
The error band obtained by computing the square root of the sum of the squares of
all applicable errors specified for the transducer.
The error band obtained by simple addition of all applicable errors specified for the